Though all of these examples are intended to serve as a starting point for the creation of a custom application, they may be used without modification to immediately control the Agilent A. Examples include VIs for reading a measured value or waveform and VIs for downloading waveforms or digital patterns to a source instrument. The returned data is described by. LabVIEW is an industry-standard graphical programming environment that can be used to quickly and easily acquire, analyze, and present data from the A. You can add the signal from one channel to the signal of the other channel this might be adding noise to a clean pulse, this might be a FM signal added with noise… The possibilities you have are huge. Legal Privacy Terms Trademark Acknowledgements. This example shows how to configure a measurement and take multiple readings.
|Date Added:||26 November 2008|
|File Size:||30.39 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
The Action VIs – initiate or terminate test and measurement operations, such as arming the trigger system or generating a stimulus.
Agilent 86100 Series Oscilloscope
Enhanced Jitter Analysis Software Accurate jitter analysis, including decomposition of jitter into its constituent components, is becoming critical as data rates increase.
Have tried using the instrument drivers. The alternative to using the NI Instrument Driver Finder is to go directly to the Instrument Driver Network and search for your instrument driver there. Back to Top 2. In this example, serial communication is labvidw. Numerous Configuration VIs can exist, depending on the particular instrument.
DCA 86100C labview examples
Legal Privacy Terms Trademark Acknowledgements. Message 1 of 8. I had some thoughts after our support 86100s yesterday that might give us a bit of a better idea as to what is going on.
What does channel add mean? Instrument drivers simplify control and reduce test program development time by eliminating the need to learn the unique programming and communication protocol for the instrument.
agilent a acquire waveform into labview – Discussion Forums – National Instruments
It sounds like you are getting waveforms from the scope though. To run this example, the user first sets the GPIB or serial communication parameters. Message 5 of 8.
Rate this document Select a Rating 1 – Poor 2 3 4 5 – Excellent. Is a screen capture software package available for the ESA Series spectrum analyzer? Many instrument drivers have llabview VI Tree VI you can use to view the entire instrument driver hierarchy.
Try using an ‘Array Size’ function to determine the number of points acquired from the device. The block diagram in Figure 4 extends the functionality of our previously created laview to discover the peaks of the dataset and then log those values to a tab-delimited text file compatible with Excel.
How can I find out when new firmware or software is released?
From the block diagram you can see that the instrument is first initialized with the Serial Configuration 8100d VISA resource name specified on the front panel. The most straightforward way to take advantage of the LabVIEW instrument driver is to run the example programs, which require only basic user configuration. There may be a command to retrieve what the sample rate is in the device.
Simply copying and past four times seems as a crude approach but maybe there is a better way of doing it. Capture deep memory waveforms and save them as a. Options For more details: Additionally, it can perform any necessary actions to place the instrument in its default power on state or in another specific state. These VIs include the majority of the instrument driver template VIs, such as reset, self-test, revision, error query, and error message. See search results instead: